Sensor: How many microns per pixel can be seen with the Dino-Lite?
To determine microns per pixel, measure the field of view at the magnification being used. For example, using the AM4113T(R4) at its highest magnification (~220x), the FOV is 1.8mm or 1800um. Take the FOV in microns and divide by the resolution of the Dino-Lite, so in this case we would divide 1800um by 1280 pixels. 1800um/1280 pixels = 1.41um/pixel. Since the Dino-Lite handheld microscopes offer variable magnification, the microns per pixel will be different with each magnification. The above method can be performed for any magnification you wish.